SDZ - type IIA electronic needle therapy instrument is integrated with modern electronic technology and the traditional meridian theory, with percutaneous electrical stimulation, electric acupuncture treatment and acupuncture point, and so on.The equipment safe operation is simple, reliable curative effect sure, widely used in hospital and family self-care
1,Low through the eu CE certification and German TUV PS ISO9002, EN46002 quality system certification.
3,Six Way output, automatic timing.
4,Low external regulated power supply, for a variety of purposes.
1, percutaneous electrical stimulation
3, frequency of ground hole (mainly for the points) : 1, the continuous wave: 1-100 hz, and 2 adjustable density wave of hydrophobic wave frequency is approximately a third of the dense wave frequency goes: output: 2 test point: all the way (output jack B) power supply: dc 9 v or external dc 9 v power consumption: less than or equal to 0.3 watts.
Clinical inside and outside, gynecology, dermatology, ophthalmology, surgery, orthopedics, such as disease hundreds kinds of each system, especially for the week of common diseases such as migraine, trigeminal neuralgia, facial nerve palsy, sciatica, toothache, scapulohumeral periarthritis, tennis elbow, contusion, muscle strain, acute stomach pain, menstrual cramps, constipation, diarrhea, renal colic, biliary colic, acute appendicitis, acute tonsillitis, banded blister all have remarkable curative effect.
A, Low output pulse waveform: asymmetric two-way pulse, continuous wave, discontinuous wave and the dilatational wave
B, Low output pulse way: output
C, Low output pulse frequency:
1. Continuous wave: 1-100 hz frequency adjustable
2. The discontinuous wave: 1-100 hz continuous wave frequency adjustable, breaking wave time is 5 seconds, wave time for 15 seconds
3. The density wave: thin wave frequency is 1/5 of the dense wave frequency, dense wave frequency is 5-100 hz is adjustable, sparse wave time is 5 seconds, dense wave time for 10 seconds, the output pulse frequency allowed error of 15%
Low output current limit is not more than 50 ma (500 load)
Low output pulse width is not more than 0.175 ms
Low power output is less than 3.5 W
Low supply voltage: DC9V
Low volume: 348 mm * 222 mm * 93 mm
Low weight: 1.5 KG
Low power: input, 220 v / 50 hz output DC9V / 150 ma
The host * 1
Output wire * 6
Needle electrode metal clip * 6 pair
Skin electrodes * 6 pairs
The power adapter * 1