If you need an English user manual, please leave a message under the order: email address. I will send it to your email address.
SDZ-II Nerve and Muscle Stimulator is combined with micro computer technique, traditional chinese medicine acupuncture and meridian theory, based on the traditional electronic acupuncture treatment instrument.lt doesn't contain acupuncture needles, but with Filiform Needle Electrode Metal Clip. Can use acupuncture needles as Electric-Acupuncture therapy, and recommend to use disposable acupuncture needles.
SDZ-II Nerve and Muscle Stimulator as a kind of TENS, EMS devices, may be used to apply low frequency pulse electrical current to stimulate the acupoints of human body through electrodes on a patient's skin to treat pain.
Hwato brand new SDZ - type II electronic needle therapy instrument with low (frequency) pulse current on the human body meridian acupoints, makes the ion distribution inside the organization change, to adjust the tension of nerve muscle tissue, promote the blood circulation around, make the body excitement or inhibit the partial victory partial failure condition to adjust, the balance analgesia antispasmodic, antiphlogistic, detumescence, promote the clinical effect of tissue regeneration.
Hwato electro-acupuncture instrument is also widely used in clinical internal and external, gynecology, dermatology, ophthalmology, ENT, orthopedics and traumatology and other system diseases, especially for common diseases such as migraine, trigeminal neuralgia, peripheral facial nerve palsy , Sciatica, toothache, frozen shoulder, tennis elbow, contusion, muscle strain, acute stomach pain, dysmenorrhea, constipation, diarrhea, renal colic, biliary colic, acute appendicitis, acute tonsillitis, and herpes zoster have significant effects .
MAIN TECHINCAL SPECIFICATIONS:
•Power: DC9V Use dry batteries or AC Adapter (input AC220V 50Hz/AC110V 60Hz, output DC9V)
•Power input: MAX 10.0VA (including AC Adapter)
•Output pulse wave: Un-symmetry bi-directional pulse wave
•Output pulse channel: six channels
Power output: MAX 0.3VA (Load 2500)
Output pulse frequency: 1-100Hz adjustable
Continuous Wave: Continuously.
Intermittent Wave: interval time between Continuous Wave groups 15 sec, duration time of Continuous Wave 5 sec.
Distant-dense Wave: frequency of distant wave is one fifth of the frequency of dense wave, the duration time of distant wave 5 secretary, the duration time of dense wave 10 sec.
Output current: <10mA (Load 250Ω)
Output DC amount: 0
Output pulse width: 0.2ms±30% (EMC TEST)
Fast operational guidelines for school official cites use: first, close to the right of the power switch, the 6 set of output "electric intensity adjustment knob counterclockwise to the left, until he heard" a "so far a, then open the power switch, timer, waveform output button, waveform lights, changing the output frequency switch on the left, to adjust the output strength finally
Warm prompt: computer, 6 groups were XianGui zero output of the knob, adjustable frequency knob to the left, if the frequency is zero, the output, there is no response
Before use, all buttons need to be turned off, otherwise the device cannot be started.
1 X SDZ-II Main machine
6 X Output wire
6 X Metal clip
6 X Self-adhesive electrode
1 X Power adapter
1 X English Manual